|Input Voltage:||AC100V-240V||Working Frequency:||4-7Mhz|
|Output Power:||200-400VA||Name:||2 In 1 4D&Ice Head HIFU|
|Base Weight:||14kg||Cart Pack Size:||50*36*61cm|
Facial 9D HIFU Beauty Machine,
7D HIFU Vaginal Tightening Equipment,
Wrinkle Removal HIFU Beauty Machine
What is 7D Focused Ultrasound?
The effect of 7d HIFU is to lift and firm the skin, improve the facial contour, eliminate wrinkles, whiten, and shape the body. The principle is to use the machine probe to emit high- energy focused ultrasound, instantly penetrate the skin, act on the epidermis, dermis, fat layer, fascia layer, stimulate the skin collagen contraction and regeneration, so as to achieve the effect of layering and anti-aging. 7d HIFU is a non-invasive anti-aging medical beauty project. It adopts the latest technology of MIFU and integrates the micro high-energy focusing
system and macro high-energy focusing system. Delivered to different levels of the skin, not only the anti-aging effect is better, but also the comfort of beauty seekers is higher. 7d HIFU uses the principle of focused ultrasound to instantly apply high-energy ultrasound to different layers of the skin, such as the epidermis, dermis, fat layer and fascia layer, to stimulate the regeneration of collagen fibers, improve the support of the skin, and lift and firm. Loose skin; at the same time, it also promotes the dissolution of some fat, so as to achieve the effect of body shaping.
What is 7D focused ultrasound?
The 7D HIFU adopts a miniature high-energy focused ultrasound system, the focus point is smaller than other 3D 4D HIFU equipment, and more
Accurately transmits high-energy focused ultrasound energy at 65~75°C to the target skin tissue layer, resulting in a thermal coagulation effect without damaging surrounding tissue. While stimulating the proliferation of collagen and elastic fibers, it greatly improves comfort, gives you a perfect V-face, and makes your skin plump, firm and elastic.
The main function
AThe main function
Anti-wrinkle, anti-aging, lifting, firming.
1. Facial anti-aging and anti-wrinkle: deep wrinkles, folds, forehead lines, nasolabial groove lifting, lip wrinkles, brow wrinkles, etc.;
2. Neck anti-wrinkle: tighten neck skin, improve double chin, thin face
3. Anti-wrinkle around the eyes: wrinkles around the eyes, crow's feet at the corners of the eyes, bags under the eyes, tightening the loose skin around the eyes;
4. Whole body anti-aging: Tighten the back, adjust the chest, shape the waist and abdomen, shape the buttocks, shape the legs, remove the butterfly sleeves, and repair the light.
Involving instrument parts: host, handle *2
Put the handle properly into the handle bracket first, and do not force the handle part directly into the buckle position when placing it, otherwise it may cause stress fracture;
2. Make sure the handle is placed properly;
3. Find the handle interface on the back side of the host;
4. Align the white arrow on the handle interface with the interface, and gently connect the wire to the interface;
Precautions: When placing the handle, first insert the connecting wire into the buckle first, and then insert the handle down, do not directly press the handle into the buckle position; Pay attention to the direction of the interface when connecting the handle, do not use brute force when the plug is not smooth, please check whether the direction is correct;
0.5 J/cm2 to 100 J/cm2
Maximum number of points for a single output focus
Points for a single row output
Operation of the instrument1. After receiving the instrument, unpack it and connect the power cord. 2. Connect the treatment handle. When connecting, note that the red dot on the handle line corresponds to the red dot on the instrument, just insert it. If the connection is not in place, the instrument will alarm. 3. Activate the power switch on the rear of the instrument
4. Then install the treatment probe on the treatment handle (the handle should be installed upwards to prevent the treatment probe from falling on the ground and being damaged)
5. Enter the operation interface and click anywhere on the screen to enter the preparation interface
Instrument common fault problems:
1. There is no display on the instrument interface, please check whether the power plug is not connected or the power switch is turned on. 2. The instrument is normal but there is no energy output or the alarm keeps on turning on. Please check
whether the connection between the handle wire
and the host is firm or whether the probe is installed in place.
Maintenance and care of the instrument
1. After using the instrument, please disinfect it with alcohol or normal saline, wipe it clean and keep it properly.
2. Before use, a plug with a grounding pin must be used, and ensure that the power socket of the instrument is truly grounded.
3. The voltage of the instrument is suitable. If the local power supply voltage is unstable, we recommend that the user add a regulated power supply with matching power.
4. For the therapeutic effect of the instrument and to ensure normal service life, please use the designated accessories provided or recommended by the original manufacturer.
5. Do not place it in a humid place or near a water source, and do not expose the instrument to direct sunlight.
6. The instrument is close to a strong heat source, because this may affect the life and normal use of the instrument.
7. Please remove all metal objects on your body first, so as to avoid unforeseen situations that may affect the curative effect.
8. Avoid bad contact. Please pay attention to the place where the skin is in contact with a certain degree of moisture. Wipe it with water-based products or water to achieve the desired effect.
9. Please turn off the power switch of the instrument when the instrument is not in use, and ensure that the main power of the instrument is turned off before use to ensure the safety of electrical products.
10. Strictly follow the instructions in the user manual to use the instrument or train the operator of the instrument.
Contact Person: Admine