|Product Name:||Ice Sculpture Board Machine||Display Screen:||10.4 Inch LCD|
|Cooling Temperature:||0℃ To -11℃||Heating Temperature:||37℃ To 45℃ Preheating For 3 Minutes|
|Input Voltage:||110V/220V||Host Size:||57(L)×34.5(W)×41.5(H)cm|
Fat Freezing Body Slimming Machine,
Cellulite Removal Body Slimming Machine,
1000W Cryo Fat Freezing Machine
Ice sculpture board machine
The machine is equipped with eight handles, and supports one handle to work independently to eight handles work simultaneously. Using a new type of integrated device refrigeration plate, non-vacuum adsorption refrigeration technology, the freezing plate is flat, can be attached to the target area, and fixed with straps, which can be widely used in many parts of the body. It is an instrument that uses selective, non-invasive freezing to reduce local fat. It originated from the research and invention of Harvard University in the United States, and has been widely used in clinical practice in the United States, Britain, Canada and other countries. Since adipocytes are sensitive to low temperature, triglycerides in fat change from liquid to solid at 5°C, crystallize and age, induce adipocyte apoptosis, but do not damage other subcutaneous cells (such as epidermal cells, black cells). Cells, dermis tissue and nerve fibers).
It is a safe and non-invasive cryo-lipolysis, which does not affect normal work, requires no surgery, no anesthesia, no drugs, and no side effects. The instrument provides an efficient 360°surround controllable cooling system, and the overall cooling of the freezer is uniform.
The ideal temperature for inducing adipocyte apoptosis is -5°C to -11°C, and the cooling energy achieves non-invasive and potent lipid-lowering. Unlike adipocyte necrosis, adipocyte apoptosis is a natural form of cell death. It is to maintain the stability of the internal environment. Cells die in an autonomous and orderly manner, effectively reducing fat cells without causing damage to surrounding tissue.
1,The instrument must use a plug with a grounding pin, and ensure that the power socket of the instrument is well grounded.
2,The power supply used should be consistent with the specified power supply value marked on the machine, otherwise the machine may not work or even burn out the main board parts of the machine.
3,Ensure that the power supply is stable and adapted. If the local power supply voltage is unstable, it is recommended that the user add a regulated power supply with matching power. Special reminder: The power cord of the socket is required to be more than 1.5 square meters.
4,When using the instrument, please keep away from the wall and keep a 30cm space around the instrument for heat dissipation.
5,The instrument is a high-precision electronic instrument, please don’t place the instrument in a high temperature and humid environment.
6,The instrument uses an LCD touch screen. When tapping, try to tap with your fingertips instead of sharp objects.
7,Each accessory is a precision device, so when it is in use and after use, care must be taken to maintain it to avoid shortening its service life.
8,After each use of the accessories, it’s recommended to clean them with saline and then dry them with a dry cloth
9,Don’t use alcohol or corrosive solvents to clean the host and handle to avoid damage.
10,When using accessories, try to handle them gently, and don’t drop them by gravity to avoid damage to the handle.
11When in use, the cord hose of the handle avoids extreme bending and damage
12,When the instrument is not in use, please turn off the power, then unplug the power plug, and place various accessories of the instrument. If possible, cover the instrument with a dust cover.
13,If it’s not used for a long time or when it is being transported, the water should be drained clean.
14,It’s strictly forbidden to disassemble and modify the equipment without authorization.
15,If the equipment fails, it should be shut down immediately, please contact us.
|Product name||Ice sculpture board machine|
|Display screen||10.4 inch LCD|
|Cooling temperature||0℃ to -11℃|
37℃ to 45℃
preheating for 3 minutes
|Air box size||66×43×76.5cm||Air box weight||13kg|
1,Installation: vertically align the working handle with the handle socket on the back of the instrument host;
Long press the lock button of the handle, then connect the freezing head to the working handle and place it on the rack.
2,Warm reminder! For the first use, you must fill up with water before starting the machine. Never work without water!
Please refer to page 5 for the method of adding water.
3,Insert the power cord into the socket at the back of the main unit, turn on the power, turn on the power switch, press the power button, and the instrument will start up immediately.
Contact Person: Admine